Choosing the right laboratory fume exhaust system is essential for ensuring safety in the lab. Fume exhaust systems capture and remove toxic and hazardous fumes, vapors, and particles, protecting both lab personnel and the environment. In this guide, we’ll discuss the different types of fume exhaust systems and how to select the most appropriate one for your laboratory.
Types of Fume Exhaust Systems
There are several types of laboratory fume exhaust systems available. These include:
1. Ducted Exhaust Systems: Ducted fume exhaust systems are the most popular type of fume exhaust. They connect to a laboratory hood and use a network of ducts to exhaust hazardous fumes to the outside of the building. The ducts can be made from galvanized steel, aluminum, or plastic, and they can be connected to a variety of exhaust fans.
2. Local Exhaust Systems: Local fume exhaust systems are used in areas where a ducted system is not feasible. These systems are typically wall or ceiling mounted and capture fumes at the source. They can be used with a variety of filters, including charcoal, HEPA, and activated carbon.
3. Non-Ventilated Exhaust Systems: Non-ventilated fume exhaust systems are designed to capture and remove fumes, vapors, and particles without venting them outside. These systems are typically used in areas with limited ventilation or when the fumes are too hazardous to be vented to the outside.
4. Portable Exhaust Systems: Portable fume exhaust systems are designed to be used in laboratories where the ventilation requirements are constantly changing. These systems are typically equipped with HEPA filters and can be easily moved from one area to another.
Factors to Consider When Choosing a Fume Exhaust System
When selecting the right fume exhaust system for your laboratory, there are several factors to consider. These include:
1. Type of Contaminants: The type of contaminants being exhausted from the lab will determine the type of fume exhaust system needed. For hazardous fumes, a ducted system is typically the best option. For non-hazardous fumes, a local or non-ventilated system may be sufficient.
2. Location: The location of the fume exhaust system should be determined prior to selection. If the system is to be installed outdoors, it must be weather-resistant. If the system is to be installed indoors, the ventilation requirements must be taken into account.
NEW TYPE Lab Ceiling Mounted Flexible Hood , Flexible Fume Extraction Arm
3. Capacity: The capacity of the fume exhaust system must be sufficient to handle the volume of fumes being exhausted. If the system is too small, it will not be able to handle the load.
4. Noise Level: The noise level of the fume exhaust system should be taken into consideration. If the system is to be installed in an area where people work or live, the noise level should be kept to a minimum.
5. Maintenance: The maintenance requirements for the fume exhaust system should be taken into account. Systems that require frequent filter changes or other maintenance tasks should be avoided.
6. Cost: The cost of the fume exhaust system should be taken into account when making a selection. A higher-priced system may offer better performance, but it may not be necessary or cost-effective for your application.
Choosing the right fume exhaust system for your laboratory is essential for ensuring safety in the lab. When selecting a fume exhaust system, it is important to consider the type of contaminants being exhausted, the location, capacity, noise level, maintenance requirements, and cost. By taking all of these factors into account, you can ensure that you select the most appropriate fume exhaust system for your laboratory.
What is a laboratory fume extractor arm? The structure and characteristics of the fume extractor arm for laboratory